What is the main difference between FlashRES-64 and FlashRES-UNI?2020-04-13
FlashRES-64 only uses ZZ in-house array to collect data. It can collect over 62k data points with 64 electrodes in one hour. This large amount of data can be used to produce much better inversion result than one produced from the data of any traditional arrays.
Can FlashRES output more than 350v?2020-04-13
In theory, the voltage between M and N (Vmn) is depending on the injected current between A and B (Iab); Iab is depending on the input voltage between A and B (Vab). So if Vab increases, Vmn will then increase. But Vab cannot be increased too high, especially for multi-core cable, this is due to ...
How heavy is the FlashRES package?2020-04-13
FlashRES only weights 5kg. With suitcase design, it is very easy for user to travel with or mobilise FlashRES unit. The accessories of FlashRES system include electrodes (10kg), cables (weight depends on length) and battery (8kg).
How to set up a FlashRES system?2020-04-13
Please refer to diagram below.
What are functions of connectors on FlashRES front panel?2020-04-13
(a) Fuse (20A) 20A fuse (b) Battery (12V) 12V battery terminal connector (c) Cable 1 (1-32) 32-core cable connector for electrodes 1 - 32 (d) Ext A / Ext B Single-core AB cable connector for high voltage survey (e) Extensions Data connection wire between two F-UNI instruments for...
What are the resolution and accuracy of FlashRES systems?2020-04-13
0.1 μV / 0.2%
Are FlashRES cables active cables?2020-04-13
Currently, there are two kinds of resistivity survey cables. One is called passive cable, another is called active cable. For passive cable, there are no electronic control parts in the cable, and all control and ADC are in the instrument. For active cable, it contains electronic parts in each el...
What is the specification of FlashRES-UNIVERSAL?2020-04-13
Please refer to our FlashRES features and specifications section.